Electric Cars and Bikes Do Not Interfere with Implanted Cardiac Devices
Author: Mayo Clinic(i) : Contact: mayoclinic.com
Published: 2013-03-10 : (Rev. 2019-03-13)
Mayo Clinic cardiac investigators study potential risk of effects of EMI on patients with implantable devices.
A Mayo Clinic study has concluded that patients with implanted cardiac devices such as pacemakers and defibrillators can safely drive or ride in an electric car without risk of electromagnetic interference (EMI).
The study, titled "Hybrid Cars and Implantable Cardioverter Defibrillators: Is It Safe" is the first of its kind to address the interaction between these devices and electric cars. It was presented at the 2013 American College of Cardiology Annual Scientific Session in San Francisco on March 9th.
In some cases, implanted devices may sense signals from electrical or magnetic objects and misinterpreted them as potential distress coming from the patient's heart.
The increasing prevalence of electric and hybrid cars, one of the fastest growing segments of the American automotive industry, prompted Mayo Clinic cardiac investigators to study the potential risk of the effects of EMI on patients with implantable devices.
Mayo Clinic researchers used implantable devices from the three major manufacturers and a 2012 Toyota Prius hybrid in the study.
Electric and magnetic fields were measured in six positions:
- From the driver's seat
- Front passenger seat
- Left and right rear seats
- In front of and behind the car from the outside
Each position was evaluated at different speeds: 30 mph, 60 mph and at variable speeds of acceleration and deceleration.
The 30 study participants with implanted devices were continuously monitored while rotating positions in the car and driving the car, with a particular focus on real-time detection of any interruption in the normal functionality of their devices.
Luis R. Scott, M.D., Cardiologist, and Fernando Tondato, M.D., Cardiology Fellow, both of Mayo Clinic in Arizona and the study's lead investigators, called for additional study.
They said that although the study results revealed that the car tested did not generate clinically relevant amounts of EMI and that the car is safe for patients with implanted devices, more research will help define and solidify the results.
"Further studies may be necessary to evaluate the interaction between implantable devices and other models of hybrid or electric cars," says Dr. Scott.
The American College of Cardiology named a poster that summarizes the study and will be presented at the conference, as a 2013 "Best Fellows In Training (FIT) Poster," which represents the top 3 percent scoring percentile of abstracts accepted for presentation at the conference.
Mayo Clinic is a nonprofit worldwide leader in medical care, research and education for people from all walks of life.
The study was funded by the Mayo Clinic Center for Translational Science Activities.
(i)Source/Reference: Mayo Clinic. Disabled World makes no warranties or representations in connection therewith. Content may have been edited for style, clarity or length.
- 1 - Ribcap Protective Hats For Fall Risk Patients : Ribcap (2017/03/25)
- 2 - Folding Portable Wheelchair Scale With Ramp : Alliance Scale, Inc. (2017/08/17)
- 3 - Thermo: Wi-Fi Connected Temporal Thermometer : Withings (2016/01/06)
- 4 - Electric Cars and Bikes Do Not Interfere with Implanted Cardiac Devices : Mayo Clinic (2013/03/10)
- 5 - Disability Products: Assistive Devices for People with Disabilities and Seniors : Thomas C. Weiss (2009/06/19)
- 6 - Stimulating The Senses With a Bubble Tube : experia-usa.com (2010/07/13)
- 7 - Sensory Equipment As Teaching Aids : experia-usa.com (2010/07/13)
• Disabled World is strictly a news and information website provided for general informational purpose only and does not constitute medical advice. Materials presented are in no way meant to be a substitute for professional medical care by a qualified practitioner, nor should they be construed as such. Any 3rd party offering or advertising on disabled-world.com does not constitute endorsement by Disabled World.
• Please report outdated or inaccurate information to us.